- برند: UNI-T
- Origin: CN(Origin)
- DIY Supplies: ELECTRICAL
- گواهینامه: CE
- سایز صفحه نمایش: 3.0 – 4.9 Inches
- مدل: UTG932E/UTG962E
- Channel: 2
- Max frequency: 30MHz
- Sampling rate: 200MS/s
- Vertical resolution: 14 bits
- Waveforms: Sine, square, pulse, ramp, noise, DC, arbitrary
- Sweep modes: Logarithmic, linear
- Power: 100~240VAC, 50Hz/60Hz
UTG900 Series Function Arbitrary Waveform Generator
The UTG900 series function arbitrary waveform generator uses direct digital synthesis technology to produce accurate and stable waveform output, with a resolution as low as 1μHz. It is an economical, high-performance, and versatile function arbitrary waveform generator. It can generate accurate, stable, pure, low-distortion output signals, convenient operation, superior technical indicators and humanized graphic display. It is a multi-purpose device that meets the needs of learning and testing and improves work efficiency.
Q: How to output the arbitrary waveform of the arbitrary waveform editor?
A: Draw an arbitrary waveform through the arbitrary waveform editor, save the arbitrary waveform to the computer, open the control panel of the signal source host computer, click the right button, and send the arbitrary waveform file to output.
Q: How to output the desired arbitrary waveform (the signal originates from a signal outside the band)?
A: Edit the desired arbitrary waveform file through the arbitrary waveform editor in the host computer of the signal source, and then load it to the arbitrary waveform path of the signal source via a U disk, and then the edited arbitrary waveform can be output.
Q: The signal source output amplitude value, the oscilloscope measurement result is twice as different?
A: Since the default output load of the signal source is 50Ω, and most oscilloscope output impedance is 1MΩ, it is necessary to adjust the output load of the signal source to high impedance. After impedance matching, the measured value of the oscilloscope is consistent with the output amplitude value of the signal source.